According to the United States Pharmacopeia and the National Formulary (USP 42-NF 37*), analysis of maltose should be carried out with a column with L58 packing material and meets following requirements. The SUGAR KS-801 confirmed the requirements were met.
System suitability requirements:
Resolution of maltose and maltotriose: ≥ 1.6
Relative standard deviation (RSD) of maltose peak area: ≤ 2.0 %
*The version at the time of the application acquisition.
Sample : 10 mg/mL each
(in H2O), 20 µL
Column : Shodex SUGAR KS-801 (8.0 mm I.D. x 300 mm) Eluent : H2O Flow rate : 0.35 mL/min Detector : RI Column temp. : 80 °C
Sample Name Index
Operation Manual / Certificate of Analysis
Operation Manuals and Certificate of Analysis / Inspection Certificate for the following products can be downloaded here.