Inositol was analyzed following the Japanese Pharmaceutical Excipients 2018. The HPLC column packed with a Na-form, strong acidic ion-exchange resin should be used. Shodex KS-801 is one of the columns.
The system suitability requires to satisfy resolution (Rs) between inositol and 1-propanol, an internal standard, to be ≥ 2.0 for a 10-µL injection of the standard solution. Also, the relative standard deviation (RSD) of inositol and 1-propanol to be ≤ 1.0 % for the six repeated analysis. It was confirmed that KS-801 satisfies all required conditions.
Sample: 10 µL
1. myo-Inositol 0.4 %
2. 1-Propanol 1.2 %
Column : Shodex SUGAR KS-801 (8.0 mm I.D. x 300 mm) Eluent : H2O Flow rate : 0.9 mL/min Detector : RI Column temp. : 65 °C
Sample Name Index
Operation Manual / Certificate of Analysis
Operation Manuals and Certificate of Analysis / Inspection Certificate for the following products can be downloaded here.