Comparison of KF-602 and KF-802
Downsized organic SEC (GPC) Column, GPC KF-602
(6.0mmID*150mm) and standard
type column KF-802
(8.0mmID*300mm) are comapred using styrene
oligomer. The required analysis time is 10 min. for KF-602 and 20 min. for KF-802.
This means the analysis time for KF-602 is about half of that for KF-802. The
solvent consumtion is 6mL (0.6mL/min x 10min.) for KF-602 and 20mL (1.0mL/min
x 20min.) for KF-802. This means the solvent consumption for KF-602 is less than
one third of that for KF-802.
Columns : Shodex GPC KF-602 (6.0mmID*150mm) x 2 : Shodex GPC KF-802 (8.0mmID*300mm) x 2
Eluent : THF : THF
Flow rate : 0.6mL/min : 1.0mL/min
Detector : Shodex RI(small cell volume) : Shodex RI
Column temp. : 40°C : 40°C